LCD panel Voltmeter – amperemeter DC 20A, 100V digital voltmeter, watt and current ampere meter 1612
1: Function of measuring electrical parameters (voltage, current, active power, energy)
2: Voltage alarm function (surge alarm threshold, backlight and voltage flashing to alarm).
3: Reset function of the consumed power.
4: When disabled, the records are saved.
5: Large LCD display (voltage, current, active power, power at the same time)
6: Backlight function.
Working voltage: 6.5 - 100VDC
Test voltage: 6.5 - 100 VDC
Rated power: 20 A / 200 W
Measurement accuracy: 1.0 degrees
Size: 89.6 * 49.6 * 24.4 mm (length * width * depth
LCD screen: 51 * 30 mm (length * width)
1. Power: test range: 0-2KW
Within 1kw, the display format is 0.0-999.9W;
1 kW or more, display format is 1000 - 2000 W.
2. Energy: test range: 0-9999kwh
Up to 10 kwh is the display format 0-9999wh;
10 kw or more, the display format is 10 - 9999 kwh.
3. Voltage: test range: 6,5 - 100 V
Display format: 6.50-99.9v
4. Current: test range: 0-20A
Display format: 0.00 - 20.00
1. Light control, Short keystroke turns backlight on or off, backlight has memory function, can store on or off when it is off.
2. Refresh views
Step 1: Long press the button until "CLr" appears on the display, then release the button.
Step 2: At this point, the power flashes to show the power restore status, if you briefly press the button again, the power value is cleared and the flashing state is restored.
Step 3: If no operation is performed within 5 seconds, the power value is not erased and leaves the reset state.
3.Set the voltage alarm threshold
Step 1: Long press the button until "SET" appears on the display, then release the button.
Step 2: When entering this state, the current value of the high voltage alarm is displayed in the voltage area, the low voltage alarm value is displayed in the current area and the last digit starts flashing. If no operation is available, you can briefly press to plus 1. within 3 seconds automatically switches the digit, from the high frequency alarm value to the low frequency alarm value, the total is six digits, the range of the alarm voltage threshold is 6.5 ~ 99.9V;
Step 3: When setup is complete, long press the button until "PASS" appears on the display, which means that you have successfully set and the setting status will automatically exit.
4.Select a measurement range (If you have a 100A/50A current short circuit, the current test measurement range may switch between 50A and 100A.) CAUTION Your meter is version 0 A - 20A
Step 1: Press the button for a long time until "Currency" appears in the power supply area, and then release the
Step 2: At this point, "100A" appears on the display of the current area, briefly press the button to switch between "100A" and "50A" to trigger the external short circuit range 100A / 75mV or 50A / 75mV, be aware that the meter can only use these two short circuit specifications.
Step 3: Long press the button to exit the setting state or after 3 seconds of inactivity, it will automatically save and exit the setting state. CAUTION Your meter is version 0 A - 20A
1.Enable manual measurement of 1-3 mm.
2. This module is suitable for indoor use, please do not use it outdoors.
3.The connected load should not exceed the rated voltage, current.